Publication Type : Journal Article
Thematic Areas : Medical Sciences
Publisher : J Arrhythm
Source : J Arrhythm, Volume 32, Issue 1, p.29-35 (2016)
Url : http://www.ncbi.nlm.nih.gov/pubmed/26949428
Campus : Kochi
School : School of Medicine
Department : Cardiology
Year : 2016
Abstract : BACKGROUND: Several past clinical studies have demonstrated that frequent and unnecessary right ventricular pacing in patients with sick sinus syndrome and compromised atrio-ventricular conduction (AVC) produces long-term adverse effects. The safety and efficacy of two pacemaker algorithms, Ventricular Intrinsic Preference™ (VIP) and Ventricular AutoCapture (VAC), were evaluated in a multi-center study in pacemaker patients./ppbMETHODS: /bWe evaluated 80 patients across 10 centers in India. Patients were enrolled within 15 days of dual chamber pacemaker (DDDR) implantation, and within 45 days thereafter were classified to either a compromised AVC (cAVC) arm or an intact AVC (iAVC) arm based on intrinsic paced/sensed (AV/PV) delays. In each arm, patients were then randomized (1:1) into the following groups: VIP OFF and VAC OFF (Control group; CG), or VIP ON and VAC ON (Treatment Group; TG). Subsequently, the AV/PV delays in the CG groups were mandatorily programmed at 180/150 ms, and to up to 350 ms in the TG groups. The percentage of right ventricular pacing (%RVp) evaluated at 12-month post-implantation follow-ups were compared between the two groups in each arm. Additionally, in-clinic time required for collecting device data was compared between patients programmed with the automated AutoCapture algorithm activated (VAC ON) vs. the manually programmed method (VAC OFF)./ppbRESULTS: /bPatients randomized to the TG with the VIP algorithm activated exhibited a significantly lower %RVp at 12 months than those in the CG in both the cAVC arm (39±41% vs. 97±3%; p=0.0004) and the iAVC arm (15±25% vs. 68±39%; p=0.0067). In-clinic time required to collect device data was less in patients with the VAC algorithm activated. No device-related adverse events were reported during the year-long study period.CONCLUSIONS: In our study cohort, the use of the VIP algorithm significantly reduced the %RVp, while the VAC algorithm reduced in-clinic time needed to collect device data.
Cite this Research Publication : R. Yadav, Jaswal, A., Chennapragada, S., Kamath, P., Hiremath, S. M. S., Kahali, D., Anand, S., Sood, N. K., Mishra, A., Makkar, J. S., and Kaul, U., “Effectiveness of Ventricular Intrinsic Preference (VIP™) and Ventricular AutoCapture (VAC) algorithms in pacemaker patients: Results of the validate study.”, J Arrhythm, vol. 32, no. 1, pp. 29-35, 2016.